# Draw the constructional detail of C type bourdon tube and explain its working.

76 views
Draw the constructional detail of C type bourdon tube and explain its working.

Constructional Diagram:

Working:

• C type bourdon tube is made up of an elliptically flattened tube bent in such a way as to produce the C shape as shown in the fig. The free end of this tube is closed or sealed and the other end (fixed end) is opened for the pressure to enter.

• The free end is connected to the pointer with the help of geared sector and pinion. Calibrated scale and pointer is provided to indicate the pressure.

• The pressure which is to be measured is applied to the bourdon tube through open end. When this pressure enters the tube, the tube tends to straighten out proportional to applied pressure.

• This causes the movement of the free end and the displacement of this end is given to the pointer through mechanical linkage i.e. geared sector and pinion.

• The pointer moves on the calibrated scale in terms of pressure. The relationship between the displacement of the free end and the applied pressure is nonlinear.

Helical Tube

Construction and working : Figure shows helical type bourdon tube. It is similar to spiral element, except it is wound in the form of helix. The displacement of the tip of a helical element is larger than the spiral element. Central shaft is installed (not shown in fig.) within a helical element and the pointer is driven by this shaft using connecting links. This mechanism transmits only the circular motion of the tip to the pointer which is directly proportional to the applied pressure.

OR

Twisted Tube

Construction and working : Twisted bourdon tube element is as shown in figure. One end of the tube is fixed and other end is free for application of the twist. As we apply the pressure at the free end, it gives rotation or creates the twist at that end. As we remove the pressure the free end try to retain its original position.

OR

Spiral Tube

Construction and working : Spiral type bourdon tube is as shown in figure. Spiral tubes are made by winding several turns of the tube with its flattened cross section in the form of spiral. When the pressure to be measured is applied to the spiral, it tends to uncoil producing relatively long movement of the tip whose displacement can be indication of applied pressure. Accuracy of this element is higher than ‘C’ type element.

Explanation : The C-shaped Bourdon tube has a hollow, elliptical cross section. It is closed at one end and is connected to the fluid pressure at the other end. When pressure is applied, its cross section becomes more circular, causing the tube to straighten out, like a garden hose when the water is first turned on, until the force of the fluid pressure is balanced by the elastic resistance of the tube material. Since the open end of the tubes anchored in a fixed position, changes in pressure move the closed end. A pointer is attached to the closed end of the tube through a linkage arm and gear and pinion assembly, which rotates the pointer around a graduated scale. Bourdon-tube pressure gauges are often classified as simplex or duplex, depending upon whether they measure one pressure or two pressures. A simplex gauge has only one Bourdon tube and measures only one pressure. The pressure gauge shown in above diagram is a simple gauge. A red hand is available on some gauges. This hand is manually positioned at the maximum operating pressure of the system or portion of the system in which the gauge is installed.

## Related questions

The pressure and velocity at the throat of a Venturi tube, measuring the flow of a liquid, are related to the upstream pressure and velocity, respectively, as follows: (A) Pressure is lower but velocity is higher (B) Pressure is higher but velocity is lower (C) Both pressure and velocity are lower (D) Pressure and velocity are identical

Draw schematic diagram of welding transformer showing constructional features of a welding transformer. Also explain its working.

Draw and explain the working principle of turbine type flow meter. Mention its Application area.

Distinguish between fire tube and water tube boilers in steam power plant.

Draw and explain construction of Faiveley type pantograph collector. Also explain its working in brief.

What is Cathode ray tube ?

what is vacuum tube?

Draw the diagram and explain the working of ferrodynamic type frequency meter.

Draw the diagram of zero crossing detector using op-amp and explain its working.

Draw the circuit diagram of crystal oscillator and explain its working.

Draw symbol of LED and explain it’s working principle.

Draw power and control circuit for definite time-limit starter for slip-ring induction motor. Explain its working.

Give comparison of Bourdon tube and Diaphragm.

Draw output characteristics of common emitter [CE] configuration and explain active, saturation and cut-off regions in detail.

Explain in detail the construction & working of He-Ne laser.

Draw and explain the working of Schmitt trigger.

Draw and explain the working principle of ultrasonic level measurement system. Also give advantages, disadvantages and applications of it.

With neat diagram explain constructional details of dynamometer type wattmeter.

Which type of flow is measured by hot wire anemometer and describe its working principle with suitable diagram.

State working principle of column type load cell and state its any four applications.

Draw diagram of class A push-pull amplifier and explain its operation.

List and explain the different counters of PLC in detail.

Draw and explain the control and power diagram for D.C. injection Braking.

Draw and explain in brief speed measurement using non-contact type transducer.

Draw and explain temperature measurement using RTD. State its advantages and disadvantages.

Draw ultrasonic flow meter and explain its construction.

List the types of ADCs. Explain working principle of any one type of ADC.

What is solenoid valve ? Explain its working with the help of diagram.

Explain working of concentrating type of solar collector.

With neat sketch explain working of Weston type synchroscope.

With a neat sketch explain working of 1 phase dynamometer type power factor meter.

Explain construction & working of attraction type M.I. instrument with neat diagram.

Explain working of Weston type frequency meter.

Explain construction & working of electrodynamometer type wattmeter.

Sketch constructional diagram of the operation of electromagnetic flow meter. State its two limitations.

Draw and describe working of electronic energy meter.

Explain the basic operation of an oscilloscope with necessary diagram for each section in detail.

Draw the circuit diagram of bridge rectifier with π filter. Draw its input and output waveform.

Draw B-H curve for magnetic material and state its nature. State the significance of hysteresis loop. Also draw the hysteresis loop for hard steel and soft steel.

Explain in brief the working principle of strain gauge. Define gauge factor and list out types of strain gauges.

Explain the construction and working of a photoelectric cell with a neat diagram.

Explain the construction and working of electronic overload relay.

Draw the block diagram of regulated DC power supply and explain the function of each block.

Explain Center-tapped full wave rectifier with the help of circuit diagram and draw input output waveforms.

Draw and explain the measurement of torque using Torque Cell